Analysis - Import Setup
Importing Your Samples There are only 9 available fields that can be imported into the program. The program only needs to set the formatting once, therefore once the formatting is complete the user will not be asked to format the file again. You can ...
Z Axis Constant Move Error
Adjust the Z Home sensor (Brass Ring) on probed holder. Make sure the magnet is aligned to within 5mm from the end or the front (where wires are) of the switch.
CS2000 Error Writing to File
Your IT department must fix the read/write permissions to your operating directory: Windows XP: C:\Program Files\CS2000 – Full access Windows 10 32 & 64 bit: C:\CS2000 – Full access Error occurs because the user account that is currently logged does ...
Calibration factor moved high
Very important that the operators change the Anhydrone at regular intervals. As the Anhydrone absorbs moisture, the occurrence that this is done is dependant on the dryness of the sample and also humidity that may enter the open tube combustion ...
Sample was heated, but result not measurable and not useable.
Make sure that the continue analysis on unstable result is checked in the global analysis settings. The results that are marked with an “X” can be output by turning on the [ok] (Bottom right column at bottom of screen) from the [Print Import\Format ...